KAJAANI SD5002 PDF
Class: Instrumentation Manufacturer: Kajaani Model No: SD Part Number: Qty: 1. Description: Automatic Sample Valve. follow-up through a communicator or Kajaani interface via Ethernet. Three operating modes: • Manually initiated. • Collection timer 24/7. STEC SEC Mass Flow Controller. ₪. METSO KAJAANI SD TRANSMIITER USED. ₪. MKS Flow Meter (N2) – Model ALS3BV.
|Published (Last):||7 September 2011|
|PDF File Size:||7.9 Mb|
|ePub File Size:||15.63 Mb|
|Price:||Free* [*Free Regsitration Required]|
Piston rotation guard Hex cap screw full thread M8x45 acid-proof DIN Fasten the sampling device to the process coupling with a clamp. Special kahaani available Select PayPal Credit at checkout to have the option to pay over time. Kajaahi the actuator This operation requires a process shut-down — the sampling device must be removed from the process.
Detach the collar part 3. Air inlet to sampling device The sampling device requires an instrument water inlet for the solenoid valve controls. Switch off the instrument air.
METSO KAJAANI SD TRANSMITTER USED | eBay
Inspect visually the condition of the sampling device and its process coupling. If this seal shows a leakage, replace it; see point 6. Hot transport water See instructions in sections 3. Connect the wires to the pressure switch part 6. Closing piston Sample flow rate from the sampler is dependent on process pressure and on the size of the flow opening. Mounting collar Support pipe, e. Instrument air to sampling device? Hot pulp ejected with high pressure may cause serious accidents.
No air must leak from the connector. This item will ship to United Statesbut the seller has not specified shipping options. Made of Titan; otherwise similar to SD Push the support tool in the special tool set in position, as shown in Fig. Kaiaani box of the sampling device. Be careful not to misplace the seals! Disconnect the ssd5002 inlet hose and sample line from the sampling device Fig. Unfollow kajaani to stop getting updates on your eBay Feed.
Check the weld seams for cracks and fractures.
KURZ 454FTB-16-HHT Single Point Insertion Thermal Mass Flow Transmitter
Please enter a valid ZIP Code. Subject to credit approval. The sampling device is mounted to the process pipeline with a mounting flange or process coupling.
Replace the solenoid valve. Check the operation of the pressure switch with a multimeter. Replacing the transport water valve See drawing E V1.
Watch list is full. Connection wire AJ 1. Requires a process shutdown — sampling device must be removed from the kajaani for this replacement!
Minimum monthly payments are required. If you detect an air flow, replace the diaphragm; see point 6. See terms – opens in a new window or tab. Please enter a number less than or equal to 1.
Format see all Format. Condition see all Condition.
Spanner gap 16 mm c Tighten nut 20 properly — make sure that the connection does not move sideways when tightening it! There are 1 items available. Disconnect this hose and feel if air flows out from the connector Fig.
Use two-pair instrumentation cable for the control cabling. The order of the wires is not important. Be careful not to press the bearing bush with too much force and too deep — it must not exert pressure on the O-ring! The end of the tool contains a screw pin.
Report item – opens in a new window or tab. Installation space required for the SD sampling device. Then weld the process coupling firmly to the hole, make sure kajaanni the seam reaches all the way through the pipe wall.